An attempt has been made to study and understand Ion funnel technology for improved detection and measurement capabilities of ions originating from small molecules. Different ways to create and study collision induced fragments have been explored and strategies to achieve various kinds of all ions MS/MS have been discussed. We observed that stability of ions in the system is largely controlled by the funnel exit DC voltage and Rf voltages of the low pressure funnel also contributes to the collision induced dissociation of the ions. An appropriate mix of in source collision energy coupled with Funnel exit DC voltage and low pressure funnel Rf voltage can induce extensive fragmentation of the ions mimicking several stages of MS/MS simultaneously.
Ion funnel, QTOF, optimization, All ion MS/MS, Pseudo MSn
Fenn, J.B., Mann, M., Meng, C.K., Wong, S.F. and Whitehouse, C.M. (1990). Electrospray ionization-principles and practice. Mass Spectrom Rev., 9: 37–70.
Gillet, L.C., Navarro, P., Tate, S, Rost, H., Selevsek, N., Reiter, L., Bonner, R. and Aebersold, R. (2012). Targeted data extraction of the MS/MS spectra generated by data independent acquisition: A new concept for consistent and accurate proteome analysis. Mol. Cell. Proteomics, 11:0111o16717.
Kelly, R.T., Tolmachev, A.V., Page, J.S., Tang, K. and Smith, R.D. (2010). The ion funnel: Theory, implementations, and applications, Mass Spectrom. Rev., 29: 294-312.
McAlister, G.C., Phanstiel, D.H., Braumbaugh, J.,Westphall, M.C. and Conn, J.S. (2011). Higher energy collision activated dissociation without a dedicated collision cell. Mol. Cell. Proteomics., 10: 0111.009456.
Meier, L., Berchtold, C., Schmid, S. and Zenobi, R. (2012). Extractive Electrospray Ionization Mass Spectrometry- Enhanced Sensitivity Using an Ion Funnel, Anal. Chem., 84: 2076–2080.
NIST: National Institute of Standards and Technology; http://webbook.nist.gov/cgi/cbook.cgi?ID=C58082&Mask=200.
Peterson. D.S., Luo, Q., Hilder, E.H., Svec, F. and Frechnet, J.M. (2004). Porous polymer monolith for surface-enhanced laser desorption/ionization time-of-flight mass spectrometry of small molecules. Rapid Comm. Mass Spectrom., 13: 1504-1512.
Shaffer S.A., Tang, K., Anderson, G.A., Prior, D.C., Udseth, H.R. and Smith, R.D. (1997). A novel ion funnel for focusing ions at elevated pressure using electrospray ionization mass spectrometry. Rapid Commun Mass Spectrom, 11: 1813–1817
Shaffer, S.A., Prior, D.C., Anderson, G.A., Udseth, H.R. and Smith, R.D. (1998). An ion funnel interface for improved ion focusing and sensitivity using electrospray ionization mass spectrometry. Anal Chem., 70: 4111–4119.
Shaffer, S.A., Tolmachev, A., Prior, D.C., Anderson, G.A., Udseth, H.R. and Smith, R.D. (1999). Characterization of a new electrodynamic ion funnel interface for electrospray ionization mass spectrometry. Anal Chem., 71: 2957–2964
Takats, Z., Wiseman, J.M., Gologan, B. and Cooks, R.G. (2004). Mass spectrometry sampling under ambient conditions with desorption electrospray ionization. Science, 306: 471–473.
Tang, K., Shvartsburg, A.A., Lee, Hak-No, Prior, D.C., Buschbach, M.A., Li, F., Tolmachev, A.V., Anderson, G.A. and Smith, D.R. (2005). High sensitivity ion mobility spectrometry/mass spectrometry using electrodynamic ion funnel interfaces. Anal. Chem., 77, 3330–3339.
This work is licensed under Attribution-NonCommercial 4.0 International (CC BY-NC 4.0) © Author (s)